Agilent. EA – EA. Programmable DC Power Supplies. Data Sheet. View at Test Equipment Depot | | Find great deals for Agilent EA DC Power Supply 30 Day. Shop with confidence on eBay!. On Sale Agilent EA. We take purchase orders and can provide special school, government and corporate pricing.
|Published (Last):||5 June 2015|
|PDF File Size:||2.77 Mb|
|ePub File Size:||6.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
The new encoder worked perfectly and made the supply work like new. Keysight BenchVue software for the PC eliminates many of the issues around bench testing.
Keysight Technologies E3640A
This product is available for shipping to United States and Canada only. Fortunately Agilent has an online shop that supplies many wear-and-tear items for their agilebt products.
Request a Calibration Quote. The picture above is the new encoder from Agilent.
The power supply started up perfectly the first time and I made sure to run the self test again just in case. After the knob is pulled off the front the dark cover needs to be removed. After cleaning the grime off the front panel and removing the marker from the case I coerced the E into running a self test which it passed without hesitation. Ships in weeks. I hope my repair can help e3604a else out if they are having similar issues with their supply.
After doing a agilenr research I found that it was a known issue for the rotary encoders in these power supplies to fail. I also made sure to order a new knob to replace to old grungy one to complete my work on the supply.
By making it simple to connect, control instruments, and automate test sequences you can quickly move past the test development phase and access results faster than ever before with just a few clicks. This lead me to believe that the only issue with this power supply was that the rotary encoder was on its last legs. Would you like this item calibrated? Now to actually get it installed.
EA 30W Power Supply, 8V, 3A or 20V, A | Keysight (formerly Agilent’s Electronic Measurement)
I made sure the holes had minimal solder before pressing the new encoder into place and soldering. Because the knob is the main method of user input other than RS or HPIBit made the power supply almost impossible to use.
Other than that there are a lot of passive components and a MOV soldered on near the output. This is actually towards the front of the power supply making it a little awkward to pick up.
Skip to content I recently bought a used Agilent power supply in a very sad condition. Accredited Calibration without Data. After removing a couple of screws the whole inside of the power supply slides out the front.
It does have two speeds ailent and louder? I recently bought a used Agilent power e3640z in a very sad condition. Accredited Calibration with Data. Request a Rental Quote. Don’t forget the Software!
Finally the board is out and the rotary encoder can be replaced! Transcat makes it easy: The whole board will then slide over to the right in this picture agiletn is easily pulled out.
A couple days later I had a new rotary encoder ready to be installed.
E3640A 30W Power Supply, 8V, 3A or 20V, 1.5A
Finally on the other side where the PCB is ahilent tab routed in the board needs to be lifted gently with a small screwdriver until it clears the white plastic tab.
If you forget to remove these the PCB on the other side will not be able to be removed.
Keysight Technologies (Agilent HP) E Series – W used or new for sale at used-line
In my case the encoder would miss steps or sometimes step in the wrong direction. Because the encoder snaps into place I had to cut the two large legs off of the old encoder before I could comfortably remove it from the board.
New Purchase Calibration Learn More Save time and money by calibrating your new instrument before it leaves our warehouse, ensuring it arrives ready to be put into service.